Here’s an article by our dear member Martin M. focusing on current and voltage sensors in terms of features, applications, and benefits. Enjoy and leave your impressions.
Majority of faults, incidents, and accidents associated with electrical equipment and systems are as a result of undervoltage and overvoltage. It is important to know the exact amount of current and voltage is a particular system and application as it enables the engineer or technician to make and implement safety-critical decisions.
In addition, knowing the amount of current or voltage in a system allows one to gauge the performance of the various subcomponents in a system.
There are two primary types of current sensors, direct and indirect sensors. The direct sensor applies Ohms Law and Halls Effect. It has an integrated coil. By placing the coil around a current-carrying conductor, there in an induced voltage, which is proportional to the current in the system. By using various amplifiers such as shunt and operational amplifiers, as well as a user-friendly interface, it is then easy to detect and measure the current in the system.
Do note that the shunt is designed to withstand short-circuit currents without any distortion to the resistance value. This allows an external, non-invasive way to measure the current. Direct sensors are used for low currents (<100A). For higher currents (100-1000A), the indirect method is used. It works on Ampere’s and Faraday’s Laws by measuring the magnetic field that surrounds a current carrying conductor.
Traditionally, current sensors were used for circuit protection and control. However, the values are presently used to monitor and enhance performance of different electronics and electrical equipment.
The various applications include the following:
- Linear and switch-mode power supplies monitoring
- Motor control
- Overload protection
- Over current protection
- Ground fault detection, amongst others
The primary function of voltage sensors is to detect and measure AC and/or DC voltage levels. When the presence of voltage is detected, the sensors provide an output in the form of analogue voltage signals, current levels, frequency and modulated frequency outputs or audible sounds.
A typical sensor features a needle or pointer, while the digital versions feature an alphanumeric interface. The sensors in use include handheld voltage sensors, DIN rail mounted sensors, and PCB mounted voltage sensors.
The voltage sensor is able to measure the presence of a voltage without making metal contact i.e. on insulated wires. A voltage sensor consists of a resistive voltage divider. The integrated resistors, embedded in a casted resin (for voltages between 1-72kV), have a low value inductance. The arrangement, mostly in a zigzag arrangement, together with the resin permittivity, results to a capacitance. This capacitance is more than any capacitance straying to the ground.
Electric principles dictate that the smaller the capacitor, the larger the voltage. Thus, when you hold the voltage sensor in your hands, and place the tip near a live conductor, you create high impedance on the tip, thus forming a capacively-coupled series circuit. The circuit is complete through you, the ground, and back to the circuit. Thus, the voltage detector sensor tip is a small capacitor that is capacively-coupled to a live voltage.
There are various applications associated with voltage sensors. The common areas of use include power demand control areas, power failure detection, safety switching, revenue metering, load control especially in motors, power quality monitoring especially near VAR compensators and energy management control systems.
- Power measurement: this is usually done on industrial equipment to measure efficiency
- DC voltage measurement
- Power quality measurement
- Earth fault measurement
- Switchgear optimization
TOP 5 Best Sellers Voltage Tester on amazon
- Designed specifically for HVAC applications
- Provides non-contact determination of AC voltage in cables, cords, circuit breakers, switches, outlets and wires, as well as AC voltage in security, entertainment, communications, environmental...
- Detects AC voltage from 12 to 1000V with visual and audible indicators
- 4:1 distance-to-spot ratio IR thermometer with user selectable Degrees Fahrenheit or Degrees Celcius from -22 to 482-Degree Fahrenheit (-30 to 250-Degree Celcius)
- Convenient laser pointer to target measurement area for temperature measurement
- Extension probe attaches to Amprobe TIC 300 Pro AC voltage detector to test for high AC voltages without touching or disconnecting the circuit
- Can detect AC voltages between 1,500V and 122,000V
- For utility, industrial, and mining applications when working with high-voltage equipment such as transmission lines, downed power lines, fuses, and load-break connectors
- Extends to 57" long
- Conforms to applicable European Conformity (CE) directives and N10140 Australian standards
- Non-contact voltage detector
- Wide range of detection; finds 80V to 275kV; one instrument fits all applications
- Powered by three standard Alkaline C cell batteries
- Self-test position ensures that all system functions and indicators are working properly by energizing the complete system
- Lightweight, minimizes sway at the end of long hot sticks
- Dual sensitivity
- LED light with 100,000 hour life
- Detects from 90 to 600 V AC
- Rated to CAT IV 600 V
- Dual levels of detection
- The MM300 multimeter measures AC/DC voltage, DC current and resistance and has a safety rating of CAT III 600V, Class 2, Double insulation
- The NCVT-1 has a bright green LED to indicate the tester is working; it changes to red and warning tones sound when voltage is detected
- NCVT-1 has a digitally-controlled ON/OFF power button and auto power-off feature to conserve and extend battery life
- Receptacle tester detects wiring configurations and indicates correct wiring, and identifies common wiring faults
- Identifies open ground, open hot, open neutral, hot/ground reversed; it does not indicate quality of ground, multiple hot wires, or a combination of defects
Thanks for reading,